|ECO||3.92 Cu ft.||BMT:MC000240||
|ECO||7.8 Cu ft.||BMT:MC000241||
|ECO||14.3 Cu ft.||BMT:MC000242||
|ECO||25 Cu ft.||BMT:MC000243||
|ECO||42.8 Cu ft.||BMT:MC0000244||
|EVO||3.92 Cu ft.||BMT:MC001663||
|EVO||7.8 Cu ft.||BMT:MC001665||
|EVO||14.3 Cu ft.||BMT:MC001667||
|EVO||25 Cu ft.||BMT:MC001669||
|EVO||42.8 Cu ft.||BMT:MC001657||
The CLIMACELL® was specifically developed for exact standards in reproducing environmental conditions with a temperature range from 0°C – 100°C (ECO: -10°C up to 100°C, EVO -20°C up to 100°C optional). Controllable Humidity from 10 – 95% in 1% increments. Applications include, stability testing, packaging, food or chemicals, pharmaceutical stability and ICH, seed germination studies, plant cell or tissue cultures and insect cultures. The Climacell offers an alternative to expensive testing chambers and testing rooms. The Fuzzy logic microprocessor allows for controllable levels of humidity. When equipped with a VIS or UV lighting system uniformity parameters for test and growth conditions is exceptional.
ECO Line Features:
EVO Line Features:
|Cu ft.||3.92, 7.84, 14.27, 24.97, 42.8|
|Min Temp||Ambient -10°C|
|Temp Uniformity||± 0.5°C to 1.5°C|
|Warranty (years)||2yrs parts, 1 yr labor|
|Weight (lbs)||242 to 617|
Note: All technical data is related to 22°C ambient temperature and +/ - 10% voltage swing (if specified). For other parameters see section: Electrical connections. Temperature & humidity variation occur in the case of consistent air flow during operation. No more than 50% of the shelf should be filled in order to facilitate uniform air circulation in the chamber
Do I need a water line to connect the Climacell to?
A direct water line connection is optional. It has a built-in reservoir that can be hand-filled, or you can connect a water supply to the steam generator via an included barbed fitting. If you are connecting a water supply, the line pressure must be less than 10 psi.
Note that any water used in the Climacell should be RO or DI water, with conductivity between 15 - 25 µS. Ultra-pure water cannot be used or it will cause corrosion. Tap water also should not be used as it may cause deposits of impurities to build up.16889 BMT USA 2020-04-19